# Electrical Measurements Questions and Answers – Performance Parameters of Digital Voltmeters

This set of Electrical Measurements & Measuring Instruments Multiple Choice Questions & Answers (MCQs) focuses on “Performance Parameters of Digital Voltmeters”.

1. Basic range of DVM is _________
a) 1 or 10 V
b) 0.1 or 1 V
c) 10 or 100 V
d) 100 or 1000 V

Explanation: In a DVM, the basic range is 1 V or 10 V. Range of a DVM can be extended from a few µV to kV. This is achieved by making use of an attenuator.

2. Accuracy of a DVM is ________
a) low
b) high
c) medium
d) zero

Explanation: Accuracy of a DVM is based on the resolution. Resolution depends on the number of digits. The more the number of digits, the higher is the accuracy. For a DVM, the accuracy is of the order of ±0.005 % of the reading.

3. Input impedance of a DVM is ________
a) low
b) zero
c) high
d) medium

Explanation: Typical input impedance of a DVM is usually very high. Typical value is of the order of 10 MΩ. This is usually done in order to reduce the loading effect.

4. In DVM the common mode rejection noise is eliminated by ________
a) increasing the signal amplitude
b) making use of a resistance
c) using a transformer
d) guarding

Explanation: In a DVM, common mode noise can be eliminated through guarding. A guard is basically a sheet metal box around the circuit. This is made available to the circuit being measured by making use of a terminal at the front panel.

5. Output of DVM needs ________
a) 4 lines
b) 2 lines
c) 8 lines
d) 16 lines

Explanation: Type of DVM used depends on the digital output lines. Four lines of BCD or a single line serial output may be provided so as to read the output from a DVM.
Sanfoundry Certification Contest of the Month is Live. 100+ Subjects. Participate Now!

6. Normal mode noise can be eliminated through ________
a) attenuation
b) filtering
c) amplification
d) a transformer

Explanation: Filters can be used to eliminate the normal mode noise signals. When used for low voltage measurement, noise present at the input can cause errors in the analog to digital converter.

7. DVM has ________
a) 1-3 digits
b) 2-5 digits
c) 3-6 digits
d) 7-10 digits

Explanation: The DVM consists of 3 to 6 digits. As the number of digits are increased the resolution also increases. Thus, the resolution of a DVM is very high.

8. What is a digitizing circuit?
a) converts digital signal to analog
b) boosts the magnitude of a digital signal
c) attenuates a digital signal
d) converts analog signal to digital

Explanation: DVM converts analog signal to digital. The process is known as digitization and the circuit used to obtain the conversion is known as a digitizing circuit. Time needed for this conversion is known as digitizing period.

9. User must wait for a stable reading.
a) True
b) False

Explanation: Since it is difficult to follow the reading visually at high reading speeds, user has to wait till a stable reading is obtained. The digitizing period depends on the maximum speed at which the value is read from the meter.

Sanfoundry Global Education & Learning Series – Electrical Measurements.

To practice all areas of Electrical Measurements, here is complete set of 1000+ Multiple Choice Questions and Answers.

If you find a mistake in question / option / answer, kindly take a screenshot and email to [email protected]

Subscribe to our Newsletters (Subject-wise). Participate in the Sanfoundry Certification contest to get free Certificate of Merit. Join our social networks below and stay updated with latest contests, videos, internships and jobs!